The portable dental ultrasonic scaler is composed of main unit,
handpiece, cable, waterpipe, tip, torque wrench, footswitch, and
The portable dental ultrasonic scaler has following features:
1.1.1 Detachable handpiece can be autoclaved under 134C and
1.1.2 Automatic frequency tracking ensures that the device always
works on the best frequency, stable and efficient performance
The portable dental ultrasonic scaler contraindications:
1) The hemophilia patient is forbidden to use this equipment
2) The patients or doctors with heart pacemaker are forbidden to
use this equipment.
3)The heart disease patient, pregnant woman and children should be
cautious to use the equipment
1.3 Equipment safety classification
1)Operating mode: Continuous operatior
2)Type of protection against electric shock: Class II
3) Degree of protection against electric shock: Type B applied part
4)Applied part of the equipment: Tip
5) Degree of protection against harmful ingress of water: Ordinary
6) Degree of protection against harmful ingress of water:
protection degree against water(used on foot switch): IPX
7) Degree of safety of application in the presence of a flammable
anesthetic mixture with air or with oxygen or nitrous oxide
Equipment can not be used in the presence of a flammable anesthetic
mixture with air or with oxygen or nitrous oxide.
The portable dental ultrasonic scaler Cleaning, dlisinfecting and
5.1Please conform to the recommendations of the manual
"Reprocessing Instructions of Cleaning, Disinfecting And
Sterilizing" delivered with your product regarding the procedure of
cleaning, disinfecting, sterilizing and packing of the
components.5.2The handpiece (detachable), scaling tips,endochuck,
torque wrench, endo wrench,LED lamp and Light conductor (the
handpiece with LED lamp) can be strilized.The handpiece
(undetachable), cable and water pipe cannot be sterilized.
a)Clean the handpiece (detachable) with compressed air before
b)Be sure that the scaling tip has been unscrewed from the
handpiece and it cannot be sterilized with others.
c)Please notice whether the outer of the handpiece is damaged
during the treatment and sterilization.Don't smear any protective
oil on the surface of handpiece.
d)There are two waterproof "O” rings at the end of the
handpiece.Please lubricate them with dental lube frequently, as
sterilization and repeated pulling and inserting will reduce their
using life.Change a new one once it is damaged or worn excessively.
e) The following sterilizing methods are forbidden:
-Boil in water.
-Dip in iodine, alcohol and glutaraldehyde.
-Bake in oven or microwave oven.
Notice: We are not responsible for any damage caused in the above